Great pride goes into manufacturing the PRO OX-100B in Aquasol Corporation’s ISO Certified Factory in the USA.
The PRO OX-100B Bluetooth Enabled Programmable Digital Oxygen Monitor’s state-of-the-art technology is patent pending enabling operators to remotely log unlimited amounts of data. Users simply pair the device with the App / Software and use the on-screen menu options. This permits users the ability to access the data interface while standing up to 100 feet (30 meters) away and observe real time data recordings.
The intelligent Color Notification Signal System displays red, amber and green for a quick visual on operator set parameters. Operators will have the access to an easy-to-use App that will track an unlimited amount of data points. Data collected from the oxygen monitor can then be stored for an indefinite period to meet the strictest quality control requirements of nuclear and other highly regulated industries.
Bluetooth Transmitting Capabilities
Unlimited Data Logging Capabilities in Real Time
Intelligent Color Notification System
Designed for Universal Voltage
Self-Calibrating and Certified
Programmable in four popular languages to satisfy international customer needs
Initial Pairing of Your Device:
Re-Pairing of Your Device:
Installing the Neoprene Extension Tubing:
To Set Language:
To Set Date and Time:
|English (IN)||Metric (MM)||English||Metric|
|P-OX/B KIT||PRO OX-100B Oxygen Monitor & Accessories Kit||11 x 9 x 4||279 x 227 x 102||4.00 lb||1814 g|
|P-OX/B KIT Euro||PRO OX-100B Oxygen Monitor & Accessories Kit with European Charger||11 x 9 x 4||279 x 227 x 102||4.00 lb||1814 g|
|P-OX Sensor||PRO OX-100B Oxygen Sensor||1 x 1||21 x 20||0.03 lb||14 g|
|P-OX Batt||PRO OX-100B 9V NiMH Battery||2 x 1 x 1||48 x 26 x 17||0.12 lb||57 g|
|P-OX Charger/US||PRO OX-100B 12V NiMH Battery Charger with US Type Plug||3 x 2 x 1||8 x 60 x 25||0.10 lb||60 g|
|P-OX Charger/EU||PRO OX-100B 12V NiMH Battery Charger with European Type Plug||3 x 2 x 1||8 x 60 x 25||0.10 lb||60 g|
|P-OX USB||PRO OX-100B USB Cable (72" or 1.8m)||4 x 4 x 1||102 x 102 x 25||0.11 lb||50 g|
If the Bluetooth software cannot find the Bluetooth monitor, check to ensure your PRO OX is on and properly charged. Ensure that your computer has a BLE received (Bluetooth Low Energy). BLE must be 4.0 or higher. If not, a Bluetooth adapter must be purchased. Bluetooth software only operates on Windows 10. You may download the Bluetooth software here.
If a unit is used very frequently (daily) it should be calibrated once a week. If it is not used frequently, it should be calibrated once every three weeks.
Calibration can be performed on a weekly basis. There are no other specific recommendations regarding maintenance besides general care of the unit such as cleaning, proper storage, etc.
Every oxygen monitor is shipped with a Calibration Certificate, Certification of Conformance, and a Test Certificate. The PRO OX-100B does not require a new certificate for every calibration. With the Auto Calibration feature on these units, calibration can be done by the user. There is no need for any unit to be calibrated by Aquasol or a third party.
The Pro OX-100B is battery operated, but can also be operated through a standalone power supply. For specifications on use, refer to section 2.1.1 of the manual. With a full-charge, the battery can supply continuous power for up to two (2) hours.
When using The Pro OX-100B Oxygen Monitor for LESS than two (2) continuous hours, ensure a fully charged battery is installed into the battery housing. Then turn the power switch to the ON position. Continue to use the monitor until operation is complete or the battery indicator displays the “low battery indicator”.
When using the Pro OX®-100B Oxygen Monitor for MORE than two (2) continuous hours be sure to turn the power switch to the OFF position. If battery is installed, remove from the housing. Then connect the standalone power supply/charger cable to the port. After this step, connect the power supply/charger plug into the nearest working outlet (110V for US type/220V for EU type). Then simply turn the power switch to the ON position. The Pro OX-100B is capable of operating indefinitely with a standalone power supply.
CAUTION: ALWAYS REMOVE THE BATTERY IF USING THE MONITOR FOR MORE THAN 2 CONTINOUS HOURS. FAILURE TO DO SO CAN OVERHEAT THE BATTERY AND SERIOUSLY DAMAGE THE PRODUCT AS WELL AS VOID THE WARRANTY.
The PRO OX-100B can operate in temperatures as low as 0°C (32°F) and as high as 50°C (122°F)
Sensors are available for purchase. Each sensor has a one year warranty. It is not recommend to keep stock of extra sensors. Sensors are readily available to ship and are lightweight.
If there is an issue starting the pump when pressing the SPOT or CONT button, the pump may be damaged. Contact customer service for an RGA authorization.
If there is no display when the toggle switch is in the ON position, either the battery is drained or the LCD is damaged.
If the battery is drained, simply recharge the battery or use the standalone power supply. Refer to section 1.2.2B of the PRO OX-100B manual for instructions.
If the screen does not light up, replace the battery.
If the LCD is damaged, the unit must be sent to Aquasol Corporation for repair. Contact customer service for an RGA authorization.
If strange symbols appear on the screen when the toggle switch is in the ON position, the display is damaged.
The unit must be repaired. Contact customer service for an RGA authorization
The keypad is damaged if the keys do not correspond with what is on the display. The unit must be sent to Aquasol Corporation for repair. Contact customer service for an RGA authorization.
Ensure the sensor is plugged into the sockets properly. After running for 30 seconds, if calibration is successful, the screen will display “CALIBRATION DONE”.
The sensor has a lifespan of approximately one year under normal operating conditions (see manual for operating temperature, proper storage guidelines, etc.).